This article has Open Peer Review reports available.
Mid- term results of stryker® scorpio plus mobile bearing total knee arthroplasty
© Kobayashi et al.; licensee BioMed Central Ltd. 2012
Received: 14 October 2011
Accepted: 16 October 2012
Published: 18 October 2012
The mobile bearing knee system was introduced to lessen contact stress on the articular bearing surface and reduce polyethylene wear. The purpose of the current study was to investigate the mid-term results of patients undergoing total knee arthroplasties (TKAs) using Scorpio Plus Mobile Bearing Knee System (Stryker, Mahwah, NJ), and compare the outcomes between patients with osteoarthritis and osteonecrosis (OA·ON group) and patients with rheumatoid arthritis (RA group).
Eight males and 58 females were followed up for a period of 4.4- 7.6 years from June 1, 2003 to December 31, 2005. There were 53 knees with osteoarthritis, 17 knees with rheumatoid arthritis, and 6 knees with osteonecrosis. Clinical and radiographic follow- up was done using The Japanese Orthopedic Association knee rating score (JOA score) and Knee Society Total Knee Arthroplasty Roentgenographic Evaluation and Scoring System.
With regard to the JOA score, there was significant improvement in both groups. The postoperative range of motion was between 0.8°and 116.8° in OA·ON group, and between 0.0° and 113.7° in RA group. There were no significant differences with the radiographic evaluation between two groups. Spontaneous dislocation of a polyethylene insert occurred in one patient, and deep infection was occurred in one patient.
There was significant improvement with regard to the clinical and radiographic results of patients undergoing TKAs using the model. The risk of polyethylene insert dislocation related to the mobile bearing TKA is a cause for concern.
Total knee arthroplasty (TKA) is a safe and effective procedure designed to improve function and relieve the pain associated with osteoarthritis (OA), rheumatoid arthritis (RA), osteonecrosis (ON), and other types of arthritis. The mobile bearing knee system was introduced to lessen contact stress on the articular bearing surface and reduce polyethylene wear. Theoretically, by allowing tibial motion relative to the polyethylene insert, the mobile bearing knee system can improve knee flexibility and kinematics .
Materials and methods
Preoperative clinical data on patient subgroups
59 OA:53, ON:6
Sex (Male/ Female)
Age at time of operation (yr)
Body height (cm)
Body weight (kg)
Body-mass index (kg/m2)
Preoperative Flexion angle (°)
Preoperative Extension angle (°)
Preoperative JOA score
Preoperative FTA (°)
Follow- up duration (yr)
A midline or lateral curved surgical incision with a midvastus approach was used. After osteophytes were removed, the anterior cruciate ligament and posterior cruciate ligament were sacrificed. The modified gap technique was used to accomplish the accurate soft- tissue balance. Briefly, the distal femoral osteotomy was perpendicular to the mechanical axis and the proximal tibial osteotomy was perpendicular to the tibial axis in the coronal and sagittal plane. After the initial bone cuts, the soft- tissue release was performed. A balancer or spacer block was used to evaluate soft- tissue balances between the distal femur and the proximal tibia or between the posterior femur and the proximal tibia at both 0°extension and 90°flexion. All components were fixed with cement with resurfacing of the patella. Full weight-bearing and range of motion (ROM) exercises were begun from postoperative day 1.
Clinical and radiographic evaluation
Preoperative and postoperative clinical evaluations were performed using the Japanese Orthopaedic Association osteoarthritis or rheumatoid arthritis knee rating score (JOA score) . In the score, pain on walking, pain on ascending and descending the stairs, ROM, and joint swelling were rated for OA and ON patients with maximum scores of 30, 25, 35, and 10 points, respectively. Similarly, pain, ROM, manual muscle testing of quadriceps, ability of walking, and ability of ascending and descending the stairs were rated for RA patients with maximum scores of 40, 12, 20, 20, and 8 points, respectively. ROM was assessed using goniometer measurement. For radiographic evaluation, preoperative and postoperative radiographs of a standing anteroposterior radiograph and a lateral radiograph were utilized to assess FTA, the angle of components .
Preoperative clinical data, preoperative and postoperative JOA score, ROM, and radiographic parameters were compared between two groups using Mann- Whitney’s U test. P- values less than 0.01 were considered statistically significant.
Postoperative clinical and radiographic results of the OA·ON group and the RA group
Postoperative Flexion angle (°)
Postoperative Extension angle (°)
Postoperative JOA score
Postoperative FTA (°)
There were no significant differences with regard to the radiographic evaluation between two groups (Table 2).
Carothers et al.  have performed a meta- analysis study of 3506 mobile- bearing TKAs. In relation to 1880 rotating platform mobile- bearing TKAs, survival rates at 15 years (96.4%), increase of the mean Knee Society Score (62.6 points), mean final flexion (116.6°), and polyethylene dislocation (1.0%) have been reported, and they have concluded that the mobile- bearing TKA had demonstrated excellent results. To compare results of a mobile- bearing TKA with those of a fixed- bearing TKA, which have provided durable long term fixation and excellent survival rates, many authors have made a direct comparison on the use of mobile- bearing and fixed- bearing TKAs in the same patients [4, 7–11]. Although Price et al.  reported that the American Knee Society Score, the Oxford Knee Score, and pain scores for the mobile- bearing TKA were slightly better than those for the fixed- bearing device, most authors have concluded that there were no differences in both clinical and radiographic results between two designs [4, 7, 9–11]. As well, a meta- analysis studies have suggested no significant differences in clinical or radiological outcomes, such as Knee Society Scores, Hospital for Special Surgery scores, ROM, and radiographic alignment between mobile- bearing and fixed- bearing TKA [12, 13].
Mid- term clinical and radiographic results of the model were equivalent to results of other types of mobile- bearing TKAs described in literatures. Although a lowered posterior lip of the polyethylene insert was considered to allow deep flexion, the average postoperative flexion angle of the model was equivalent to those of other types of mobile bearing TKAs [2, 9, 13]. Postoperative flexion angle is supposed to have a good correlation with preoperative flexion angle. Additionally, it is influenced by a variety of factors, such as implant design and operative techniques. Therefore, it might be difficult to judge how much the model contributed to postoperative flexion angle. As to this model, only biomechanical laboratory study on human cadaveric specimens to assess cortex strain on the anteromedial aspect of the proximal tibia and tibial torsion from axial and rotational loading has suggested that the mobile- bearing prosthesis potentially reduced torque in the proximal tibia during knee rotation compared with the fixed- bearing prosthesis .
Dislocation of the polyethylene insert after Scorpio mobile-bearing TKA has been described previously . In general, malpositioned components, flexion-extension gap mismatch, and laxity of soft tissue tension between the femoral and tibial surfaces can cause dislocation of a mobile bearing polyethylene insert [16, 17]. In our case, the tension and alignment was considered to be adequate. Our case and duplicated saw bone model demonstrated that failure of the locking system resulted in the dislocation of the insert . Dislocation of polyethylene insert is rare and it can easily be overlooked at the early stage [18, 19]. Severe damage to either the femoral or tibial component might occur if there was delay in treatment. When a patient who has undergone TKA complains of pain, swelling, sudden instability, clicking, or locking sensation, it is important to consider insert dislocation as a possible explanation for these symptoms.
Sledge et al.  have reported that patients with RA achieve outcomes similar to those of patients with OA with regard to range of motion and pain relief after TKA. Similarly, we found no significant differences between two groups with regard to the clinical results and the radiographic results, although there was significant improvement of the JOA score in both groups. The Norwegian Arthroplasty Register  have shown that the cumulative 5- year survival of TKAs was 98.9% in RA patients and 99.3% in OA patients, with revision for infection as the end point. It means that the risk of revision of primary TKAs was statistically higher in RA patients than in OA patients. Therefore, longer follow- up evaluation is necessary to compare results of OA patients and RA patients.
There are some limitations to this study. First, the follow up rate is not adequate for the mid- term results. Although improved follow- up rate might bring different clinical and radiographic results, the number of cases is supposed to be still adequate. Second, there are no control groups in this study. We should have compare results of the design with those of other types of a mobile- bearing TKA.
In conclusion, there was significant improvement with regard to the clinical and radiographic results of the Scorpio mobile- bearing knee model as well as other types of mobile TKA. The risk of polyethylene insert dislocation related to the mobile bearing TKA is a cause for concern.
The authors report no conflict of interest.
- Buechel FF, Pappas MJ: New Jersey low contact stress knee replacement system. Orthop Clin North Am. 1989, 20: 147-177.PubMedGoogle Scholar
- Carothers JT, Kim RH, Dennis DA, Southworth C: Mobile-Bearing Total Knee Arthroplasty: A Meta- Analysis. J Arthroplasty. 2011, 26 (4): 537-542. 10.1016/j.arth.2010.05.015.View ArticlePubMedGoogle Scholar
- Matsuda S, Mizu-uchi H, Fukagawa S, Miura H, Okazaki K, Matsuda H, Iwamoto Y: Mobile-bearing prosthesis did not improve mid-term clinical results of total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2010, 18: 1311-1316. 10.1007/s00167-010-1143-z.View ArticlePubMedGoogle Scholar
- Watanabe T, Tomita T, Fujii M, Hashimoto J, Sugamoto K, Yoshikawa H: Comparison between mobile-bearing and fixed-bearing knees in bilateral total knee replacements. Int Orthop. 2005, 29: 179-181. 10.1007/s00264-005-0646-6.View ArticlePubMedPubMed CentralGoogle Scholar
- The Japanese Orthopaedic Association Japanese Knee Society: Assessment criteria for knee diseases and treatments. 1994, Tokyo: KaneharaGoogle Scholar
- Ewald FC: The Knee Society Total Knee Arthroplasty Roentgenographic Evaluation and Scoring System. Clin Orthop Relat Res. 1989, 248: 9-12.PubMedGoogle Scholar
- Kim YH, Kook HK, Kim JS: Comparison of Fixed-Bearing and Mobile-Bearing Total Knee Arthroplasties. Clin Orthop Relat Res. 2001, 392: 101-115.View ArticlePubMedGoogle Scholar
- Price AJ, Rees JL, Beard D, Juszczak E, Carter S, White S, de Steiger R, Dodd CAF, Gibbons M, McLardy-Smith P, Goodfellow JW, Murray DW: A mobile-bearing total knee prosthesis compared with a fixed-bearing prosthesis. J Bone Joint Surg Br. 2003, 85 (1): 62-67. 10.1302/0301-620X.85B1.13233.View ArticlePubMedGoogle Scholar
- Kim YH, Kim JS: Prevalence of Osteolysis After Simultaneous Bilateral Fixed- and Mobile- Bearing Total Knee Arthroplasties in Young Patients. J Arthroplasty. 2009, 24 (6): 932-940. 10.1016/j.arth.2008.05.005.View ArticlePubMedGoogle Scholar
- Hasegawa M, Sudo A, Uchida A: Staged bilateral mobile-bearing and fixed-bearing total knee arthroplasty in the same patients: a prospective comparison of a posterior-stabilized prosthesis. Knee Surg Sports Traumatol Arthrosc. 2009, 17: 237-243. 10.1007/s00167-008-0662-3.View ArticlePubMedGoogle Scholar
- Kim TK, Chang CB, Kang YG, Chung BJ, Cho HJ, Seong SC: Early clinical outcomes of floating platform mobile-bearing TKA: longitudinal comparison with fixed-bearing TKA. Knee Surg Sports Traumatol Arthrosc. 2010, 18 (7): 879-888. 10.1007/s00167-009-0985-8.View ArticlePubMedGoogle Scholar
- Smith TO, Ejtehadi F, Nichols R, Davies L, Donell ST, Hing CB: Clinical and radiological outcomes of fixed- versus mobile-bearing total knee replacement: a meta-analysis. Knee Surg Sports Traumatol Arthrosc. 2010, 18: 325-340. 10.1007/s00167-009-0909-7.View ArticlePubMedGoogle Scholar
- Smith H, Jan M, Mahomed NN, Davey JR, Gandhi R: Meta-Analysis and Systematic Review of Clinical Outcomes Comparing Mobile Bearing and Fixed Bearing Total Knee Arthroplasty. J Arthroplasty. 2011, 26 (8): 1205-1213. 10.1016/j.arth.2010.12.017.View ArticlePubMedGoogle Scholar
- Bottlang M, Erne OK, Lacatusu E, Sommers MB, Kessler O: A Mobile-bearing Knee Prosthesis Can Reduce Strain at the Proximal Tibia. Clin Orthop Relat Res. 2006, 447: 105-111.View ArticlePubMedGoogle Scholar
- Kobayashi H, Akamatsu Y, Taki N, Ota H, Mitsugi N, Saito T: Spontaneous dislocation of a mobile-bearing polyethylene insert after postrerior-stabilized rotating platform total knee arthroplasty: A case report. Knee. 2011, 18: 496-498. 10.1016/j.knee.2010.10.007.View ArticlePubMedGoogle Scholar
- Huang CH, Ma HM, Liau JJ, Ho FY, Cheng CK: Late dislocation of rotating platform in New Jersey Low-Contact Stress knee prosthesis. Clin Orthop Relat Res. 2002, 405: 189-194.View ArticlePubMedGoogle Scholar
- Thompson NW, Wilson DS, Cran GW, Beverland DE, Stiehl JB: Dislocation of the rotating platform after low contact stress total knee arthroplasty. Clin Orthop Relat Res. 2004, 425: 207-211.View ArticlePubMedGoogle Scholar
- Poulter RJ, Ashworth MJ: A case of dissociation of polyethylene from its metal baseplate in a "one piece" compression-moulded AGC tibial component. Knee. 2005, 12: 243-244. 10.1016/j.knee.2004.08.007.View ArticlePubMedGoogle Scholar
- Hedlundh U, Andersson M, Enskog L, Gedin P: Traumatic late dissociation of the polyethylene articulating surface in a total knee arthroplasty—a -case report. Acta Orthop Scand. 2000, 71: 532-533. 10.1080/000164700317381298.View ArticlePubMedGoogle Scholar
- Sledge CB, Walker PS: Total Knee Arthroplasty in Rheumatoid Arthritis. Clin Orthop Relat Res. 1984, 182: 127-136.PubMedGoogle Scholar
- Schrama JC, Espehaug B, Hallan G, Engesaeter LB, Furnes O, Havelin LI, Fevang BT: Risk of Revision for Infection in Primary Total Hip and Knee Arthroplasty in Patients With Rheumatoid Arthritis Compared With Osteoarthritis: A Prospective, Population-Based Study on 108,786 Hip and Knee Joint Arthroplasties From the Norwegian Arthroplasty Register. Arthritis Care Res. 2010, 62 (4): 473-479. 10.1002/acr.20036.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.