This article has Open Peer Review reports available.
Simple method for confirming tibial osteotomy during total knee arthroplasty
© Mutsuzaki and Ikeda; licensee BioMed Central Ltd. 2012
Received: 17 December 2011
Accepted: 31 October 2012
Published: 15 November 2012
Achieving precise implant alignment is crucial for producing good outcomes after total knee arthroplasty (TKA). We introduce a simple method for confirming the accuracy of tibial osteotomy during TKA.
Two metallic markers were placed on the skin 20 cm apart, one on the tibial tuberosity and other on the tibial crest, points that are easily identified and palpated intraoperatively. Anteroposterior radiographs of the legs were obtained. We defined the line along the markers as the tuberosity line. The osteotomy line is perpendicular to the anatomical axis of the tibia. We then calculated the angle between these two lines and designated it the osteotomy angle. We set the osteotomy angle of the protractor, and cut the bone parallel to the osteotomy line of the protractor. Postoperatively, we analyzed the varus angle of the tibial osteotomy in 35 TKAs using the protractor. The average of the varus angle of the tibial osteotomy was 89.4° ± 1.6° (95% confidence interval of −1.0976, 0.0119). There was no significant difference from the target angle of 90° (p = 0.055). The varus angles of 90° and 90° ± 2° for the tibial osteotomy were 42.9% and 82.9%, respectively.
We determined the accuracy of the tibial osteotomy in the coronal plane using the protractor to be satisfactory.
Correct alignment of the tibial and femoral components is one of the most important factors determining favorable long-term results of a total knee arthroplasty (TKA). It is generally accepted that the tibial component should be placed perpendicular to the anatomical axis of the tibia [1–3]. We use extramedullary guides for cutting the tibial bone because they help us avoid the potential complications of intramedullary guide use, including fat embolization, intraoperative fracture, and inability of intramedullary rod passage due to deformity . Placement of the tibial component in excessive varus is the commonest cause of suboptimal alignment using extramedullary guides . Coull et al.  showed that 48% of TKAs had tibial component angles of < 87° even when they tried to cut the tibia perpendicular to its anatomical axis. Also, there was a significant decrease in the probability if the implant was inserted in any varus alignment . Aglietti and Buzzi  demonstrated that varus alignment of the knee and a varus tilt of more than 2° of the tibial component correlated with the incidence of lucent lines around the tibial implant. Computer-assisted navigation, digital preoperative planning and patient-specific guides improved accuracy of tibial osteotomy in TKAs [9–12]. Recently, bone-cutting devices have improved accuracy of implant positioning, even if TKA is performed manually or using minimally-invasive technique [13, 14].
We developed our original intraoperative protractor to place on top of a tibial tuberosity and tibial crest so that we can easily palpate from the body’s surface intraoperatively to confirm that the tibial osteotomy has obtained the desired tibial angle. Therefore, we hypothesized that it is possible to cut a bone precisely using a protractor intraoperatively. The purpose of this study was to evaluate the accuracy of the tibial bone cut angle using an intraoperative protractor during TKA.
Patients and methods
From October 2000 to April 2002, a total of 3 men and 24 women (aged 23–87 years, mean 61.1 years) underwent 35 TKAs for osteoarthritis (n = 14) or rheumatoid arthritis (n = 21) using the Scorpio nonrestrictive geometry cruciate retaining system (Stryker Howmedica Osteonics; Allendale, NJ, USA). The ethics committee of Ichihara Hospital reviewed and approved the study (the reference number for the ethics approval: 1203, and the trial registration number: 1203). Informed consent was obtained from each patient. A single surgeon performed all of the operations. They were done with the patient under epidural or general anesthesia and using the medial parapatellar approach. The target alignment of tibia osteotomy was perpendicular to the anatomical axis of the tibia. We used our original intraoperative protractor and an extramedullary guide.
Intraoperative protractor method
The data for the varus angle of the tibial osteotomy was compared with the target angle (90°) using Student’s t-test, with p < 0.05 indicating significance.
The average varus angle of the tibial osteotomy using the protractor was 89.4° ± 1.6° [95% confidence interval (CI) was −1.0976, 0.0119]. There was no significantly difference from 90° (p = 0.055). The 90° and the 90° ± 2° tibial osteotomy varus angles were 42.9% and 82.9%, respectively.
The average varus angle of the tibial osteotomies using the protractor was 89.4° ± 1.6° (95% CI was −1.0976, 0.0119). The intraoperative protractor method is simple, and the accuracy is high. The tibial tuberosity and the tibial crest are easily palpated, so it is easy to identify the tuberosity line during the operation. The plasticity of the intraoperative tuberosity line can be high. The protractor can be used to confirm the cutting line after setting osteotomy jigs as an indicator. Recently, the use of a computer-assisted navigation system that achieved a high degree of accuracy relative to the desired target alignment has been described [9, 10, 14]. However, unpredictable complications, such as displaced or stress femoral or tibial fractures, have been reported to occur a few weeks after the operation with the use of a computer navigation system [15, 16]. The protractor method is simple and low-priced, and the accuracy is high.
There were no complications using the protractor method. However, there are a few cautions to be noted. Anteroposterior radiography of the tibia is necessary before the operation. Also, the tibial tuberosity must be well palpated to reproduce the position of the metallic marker before the surgery: 5 mm away from the tibial tuberosity can make the error of 1.5° when performing the osteotomy. If there is deformity in the distal tibial crest where the marker has been placed, the tibial osteotomy may not be perpendicular to the tibial shaft. With these precautions, we found satisfactory accuracy for performing tibial osteotomy in the coronal plane using the protractor.
Regarding the limitation of this study, the present method was just 2-dimensional (2D), and accuracy of implant positioning was also performed with 2D radiographic analysis, which was far behind the current 3-dimensional (3D) tools, such as computed tomography (CT)-based navigation [9, 10, 14], 3D-preoperative planning software , and patient specific instrumentation system . Therefore, a potential weakness of the study was the lack of spatial recognition. However, the present method without CT appears to have a merit in minimizing the risk of radiation exposure compared to the current tools. The study lacked a control group without the protractor, and we used small numbers of subjects. The comparative study between using the protractor and a conventional method using large numbers of subjects are necessary to clarify the effect of the protractor in the future.
The authors thank Iso Medical Systems Inc., Tokyo, Japan for technical support.
- Insall JN, Binazzi R, Soudry M, Mestriner LA: Total knee arthroplasty. Clin Orthop. 1985, 192: 13-22.PubMedGoogle Scholar
- Bargren JH, Blaha JD, Freeman MAR: Alignment in total knee arthroplasty. Corr elated biomechanical and clinical observations. Clin Orthop. 1983, 173: 178-183.PubMedGoogle Scholar
- Scuderi GR, Insall JN: The posterior stabilised knee prosthesis. Orthop Clin North Am. 1989, 20: 71-78.PubMedGoogle Scholar
- Dennis DA, Channer M, Susman MH, Stringer EA: Intramedullary versus extramedullary tibial alignment systems in total knee arthroplasty. J Arthroplasty. 1993, 8: 43-47. 10.1016/S0883-5403(06)80106-3.View ArticlePubMedGoogle Scholar
- Bankes MJK, Back DL, Cannon SR, Briggs TWR: The effect of component malalignment on the clinical and radiological outcome of the Kinemax total knee replacement. Knee. 2003, 10: 55-60. 10.1016/S0968-0160(02)00050-9.View ArticlePubMedGoogle Scholar
- Coull R, Bankes MJK, Rossouw DJ: Evaluation of tibial component angles in 79 consecutive total knee arthroplasties. Knee. 2003, 6: 235-237.View ArticleGoogle Scholar
- Lewallen DG, Bryan RS, Peterson LF: Polycentric total knee arthroplasty. A ten-year follow-up study. J Bone Joint Surg Am. 1984, 66: 1211-1218.PubMedGoogle Scholar
- Aglietti P, Buzzi R: Posteriorly stabilised total-condylar knee replacement. Three to eight years’ follow-up of 85 knees. J Bone Joint Surg Br. 1988, 70: 211-216.PubMedGoogle Scholar
- Kuo CC, Bosque J, Meehan JP, Jamali AA: Computer-assisted navigation of total knee arthroplasty for osteoarthritis in a patient with severe posttraumatic femoral deformity. J Arthroplasty. 2011, 26: 976-e17-20View ArticlePubMedGoogle Scholar
- Ishida K, Matsumoto T, Tsumura N, Kubo S, Kitagawa A, Chin T, Iguchi T, Kurosaka M, Kuroda R: Mid-term outcomes of computer-assisted total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2011, 19: 1107-1112. 10.1007/s00167-010-1361-4.View ArticlePubMedGoogle Scholar
- Vanin N, Kenaway M, Panzica M, Jagodzinski M, Meller R, Krettek C, Hankemeier S: Accuracy of digital preoperative planning for total knee arthroplasty. Technol Health Care. 2010, 18: 335-340.PubMedGoogle Scholar
- Nunley RM, Ellison BS, Zhu J, Ruh EL, Howell SM, Barrack RL: Do patient-specific guides improve coronal alignment in total knee arthroplasty?. Clin Orthop Relat Res. 2012, 470: 895-902. 10.1007/s11999-011-2222-2.View ArticlePubMedGoogle Scholar
- Seo JG, Moon YW, Park SH, Shim JW, Kim SM: An alternative method to create extramedullary references in total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2012, 20: 1339-1348. 10.1007/s00167-011-1668-9.View ArticlePubMedGoogle Scholar
- Hasegawa M, Yoshida K, Wakabayashi H, Sudo A: Minimally invasive total knee arthroplasty: comparison of jig-based technique versus computer navigation for clinical and alignment outcome. Knee Surg Sports Traumatol Arthrosc. 2011, 19: 904-910. 10.1007/s00167-010-1253-7.View ArticlePubMedGoogle Scholar
- Massai F, Conteduca F, Vadalà A, Iorio R, Basiglini L, Ferretti A: Tibial stress fracture after computer-navigated total knee arthroplasty. J Orthop Traumatol. 2010, 11: 123-127. 10.1007/s10195-010-0096-9.View ArticlePubMedPubMed CentralGoogle Scholar
- Beldame J, Boisrenoult P, Beaufils P: Pin track induced fractures around computer-assisted TKA. Orthop Traumatol Surg Res. 2010, 96: 249-255. 10.1016/j.otsr.2009.12.005.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.